Share this Article
Common Insulin Pump Features
When you’re in the market for a new insulin pump, the choices can seem overwhelming. It may help to know that the different insulin pump manufacturers and models have more in common than you may think. Here is a summary focusing on the similar features offered by many of the different insulin pumps available in Canada.
The following features are available on most insulin pumps:
1. Extremely small basal insulin increments
This function refers to the background (“basal”) insulin, which is delivered every few minutes via the insulin pump. In an injected insulin program, “basal” insulin includes long-acting insulin (ex. Lantus, Levemir) or intermediate-acting insulin (ex. N/NPH); however, in an insulin pump program, these are replaced by rapid-acting insulin delivered in very small amounts every few minutes; the rate at which it is delivered is called the “basal rate”, and is expressed in units of insulin per hour (U/hr). All of the pumps currently available in Canada have basal rates of 0.025 to 0.05 U/hr.
Small increments allow you to finely tune basal rates for individual needs, and are especially useful for small children and others who are particularly insulin-sensitive. To put this into perspective, a basal rate of 0.025u/hr means that 25/1000ths of a unit are being delivered every hour. If a portion of this hourly amount is being delivered every 3 minutes (for 20 portions in 60 minutes), for example, then just over 1/1000th of a unit (!) is delivered in each dose. That’s 400 times smaller than the smallest dose you can deliver using an insulin pen! These teeny tiny increments in dosing allow high precision when programming for basal insulin rates.
2. Alternate Basal Pattern programming
Most pumps give you the option of switching between different Basal Programs: you can set different rates for basal insulin delivery to adjust for anticipated differences in activity level (ex. school days versus weekends; workout days vs. non-workout days). For example, if your child is very active in school sports during the week, but spends most of her time on the weekends texting her friends as she lounges on her bed, she may need less insulin during the school week. If, however, your child is part of the chess club at school, but plays soccer and hockey on the weekend, she may need less insulin on the weekends than during the school week.
3. Temporary Basal Rate programming
When your child’s insulin needs are less or more than usual due to a temporary cause (ex. illness; particularly active, or inactive, days; consumption of high fat foods), you can set a temporary basal rate (“temp basal”) which either reduces or increases the amount of insulin delivered for a specified number of hours (ex. 10% less insulin for 8 hours; 30% more insulin for 3 hours).
4. Blood glucose meters that communicate blood glucose levels to the insulin pump
If you use the associated meter to check blood glucose, you don’t have to re-enter the reading into the insulin pump itself – it will send the information wirelessly, to be used in bolus calculations and for record-keeping. Alternatively, you can use a non-associated meter if you like (ex. a mini-meter in his pocket when your son goes skate-boarding for the morning), and then enter the blood glucose info into the insulin pump as needed.
5. Insulin Dose (Bolus) Calculator
Once you program in an insulin-to-carbohydrate ratio (I:C) of 1:10, for example (that is, you tell the insulin pump that your child needs 1 unit of insulin for every 10g of carbs) the bolus calculator will do the math for you, giving a suggested insulin dose for the number of carbs you dial-in for a particular meal. For example, for the above insulin-to-carbohydrate ratio (I:C) of 1:10, if your child plans to eat 52g of carbs at lunch, it will suggest a dose of 5.2U of insulin.
This may not seem all that useful given a nice round number like 10… but what if you’re dealing with 18 and 41? Or 4 and 91?
In this case, the math becomes more complex; the bolus calculator becomes more and more handy the less and less “round” your numbers are. It’s also very useful for young children or individuals with special needs to operate their insulin pump independently – they don’t have to be able to do the math at all, they just have to know how many carbs are in the food in front of them.
In the same way, the pump bolus calculator will also do the math for corrections of above-target (and below-target) blood glucose, based on your programmed Insulin Sensitivity Factor (ISF). The pump will do the math for you to calculate the units of insulin needed for a given blood sugar reading, based on the ISF you have programmed in.
At meal times, it will also combine these two recommended doses (for carbs and for BG correction), presenting you with a recommendation for the total dose.
6. Variable Insulin Sensitivity Factors (ISF)
Correction Factors or Insulin Sensitivity Factors (ISF) (that is, the amount of insulin given to treat high blood glucose) can vary at different times of the day, with the amount adjustable by the user.
7. Active Insulin calculation (Insulin on Board, Bolus on Board, Active Insulin Remaining)
With the Active Insulin Remaining function, the pump will calculate how much insulin is still acting from the last bolus for food or for high blood glucose. Certain models, under certain circumstances (ex. below-target blood glucose reading) may subtract that amount from the current bolus recommendation (which may prevent a low later).
8. Extended Bolus (Combo Bolus; Dual Wave Bolus)
An Extended Bolus (also known as a Square Wave, Dual Wave or Combo Bolus) allows you to spread out the total bolus over time. This comes in handy if you eat a food that digests slowly, or if you’re grazing at a party, or if you eat “tricky” foods like pizza.
9. Bolus shortcut buttons
Insulin can be delivered in pre-set increments at the push of a button, allowing quick and discrete insulin delivery, as well as insulin delivery for individuals with compromised vision, who may have trouble reading the pump screen.
10. Bolus Reminder alarms
Missed Bolus Reminder alerts the user if a bolus is not given within a user-set timeframe. For example, if a bolus was not given within an hour after 12:00pm, your child’s usual lunch time, an Alert will ring, with a message reminding them to bolus if necessary.
11. Pump History
Insulin pumps keep track of every bit of insulin delivered and when, as well as glucose readings entered manually or shared from a linked meter or Continuous Glucose Monitor. (Plus things like Total Daily Dose, and other summary data.) This empowers you to look back and get the facts, so you can figure out what went wrong when things don’t go as planned. It also provides the info needed to decide what to do now. The pump’s history is a great safety and glucose management tool.
12. Other common insulin pump features
- Maximum Settings for basal rates, bolus amounts, and total daily insulin allow you to set the maximum amount of insulin that the insulin pump will allow a user to deliver as basal insulin, in a given bolus, and across the whole day. This can be helpful in preventing accidental over-delivery of insulin, especially by young children.
- Auto shut off option will turn the insulin pump off automatically if no buttons are pressed within a specified time range of 1-24 hours. Or you can turn this function off.
- Suspend and resume functions stop all insulin delivery and then restart, respectively, at the push of a button.
- Low reservoir warning gives you a heads up that the insulin pump is starting to run low on insulin.
- Locking keypad prevents insulin delivery by an unauthorized user, such as a young child.
- Most pump models offer multiple types of infusion sets (short/long cannula; angled or 90 insertion; flexible or metal cannula; with or without auto-inserter device).
Although it seems most insulin pump users have a favourite, there is no such thing as the “best” insulin pump in absolute terms; the best insulin pump for your child is the one that has features that fit your family’s values and lifestyle, and minimizes the features that you don’t find desirable. With that in mind, all the best on your pump hunt!
The above information was reviewed for content accuracy by clinical staff of the Alberta Children’s Hospital Diabetes Clinic.
The above information was reviewed for content accuracy by representatives of Animas Canada and Medtronic Canada.
This material has been developed from sources that we believe are accurate, however, as the field of medicine (in particular as it applies to diabetes) is rapidly evolving, the information should not be relied upon, as it is designed for informational purposes only. It should not be used in place of medical advice, instruction and/or treatment. If you have specific questions, please consult your doctor or appropriate health care professional.
Share this Article